Published: 23 February 2015
23 February 2015
Invacare is recalling selected Power Wheelchair models manufactured between specific dates to replace the joystick or driver controls.
|SPJ+ Joysticks||Manufactured between 9 September 2009 and 21 March 2013|
|MK6i controls (excluding all ASL, Sip-n-Puff and Switch control systems)||Manufactured between 1 October 2006 and 21 March 2013|
Serial numbers of affected wheelchairs fall within the range 06Jxxxxxxx to 13Cxxxxxxx. The first two digits represent the year, the first letter represents the month (A=January, B=February, etc.).
A total of 1775 affected units have been supplied in New Zealand.
The manufacturer has identified a potential wiring defect in the joystick of a small number of wheelchairs which may result in the following actions:
Should this occur, users are advised to immediately release the joystick to cause the power wheelchair to slow to a stop, as per the instruction manual.
Users whose contact details were known by the wheelchair funders, or Invacare resellers, were contacted previously. A significant number of users of affected power wheelchairs were not able to be identified or contacted about this action. Invacare has now issued a public recall notice to find and correct remaining affected power wheelchairs.
If you believe your Invacare Power Wheelchair is impacted by this recall action, and you have not already been contacted about this issue, please call Invacare New Zealand on 0800 468 222 or email@example.com to determine if your wheelchair is one of the affected units.
Additional information about this matter is published on the Invacare New Zealand website.